Autori: Adriana Valcu
Editorial: Prof. Sergey Y. Yurish, Sensors & Transducers Journal,(ISSN 1726-5479), Vol.132, Issue 9, 2011.
Based on the reference document , the article proposes the way to calculate the errors of indication and associated measurement uncertainties, by resorting to the general information provided by the calibration certificate of a balance (non-automatic weighing instruments, shortly NAWI) used in medical field.
The paper may be also considered a useful guideline for: operators working in laboratories accredited in medical (or other various fields) where the weighing operations are part of their testing activities; test houses, laboratories, or manufacturers using calibrated non-automatic weighing instruments for measurements relevant for the quality of production subject to QM requirements (e.g. ISO 9000 series, ISO 10012, ISO/IEC 17025); bodies accrediting laboratories; accredited laboratories for the calibration of NAWI.
Article refers only to electronic weighing instruments having maximum capacity up to 30 kg. Starting from the results provided by a calibration certificate it is presented an example of calculation.
Cuvinte cheie: ISSN 1726-5479, research, development and application of sensors and transducersa,IFSA